Volume 12 Supplement 1
Relating thalamic neuronal activity and EMG for validating predictive control of deep-brain stimulation in Essential Tremor patients
© Graupe et al; licensee BioMed Central Ltd. 2011
Published: 18 July 2011
Neuronal activity at the VIM nucleus of the thalamus was recorded in Essential Tremor (ET) patients during implantation of deep-brain stimulation (DBS) electrodes and compared with surface-EMG (sEMG) taken both during implantation and at later outpatient sessions at University of Illinois Hospital, Chicago. The goal of these studies was to investigate if it is possible to use sEMG signals for predicting onset of tremor and consequently whether predictive EMG-control of DBS in ET patients does or does not contradict predictive features of neuronal activity in the brain. For this purpose, we examined spike-rate and local field potentials (LFP) at and in the vicinity of VIM. Specifically, we compared spike-rate, LFP and sEMG recorded before versus after applying short DBS pulse-trains during implantation surgery, as well as sEMG recorded from limbs or neck of the patients. Out of 4 ET patients involved in the tests, we have implant-session data for 3 patients. Three patients had follow-up EMG testing.
All neuronal activity results (patients ET1, ET3, ET4) indicate sharp drops in thalamic spike rate at cessation of a simulation pulse train of 15-40 seconds as was the case in one of the two patients were LFP power was measured. The delays were all of the order of the delays (12-40 sec.) that were found in sEMG parameters at implantation sessions and at follow up session, and which we successfully used to predict onset of tremor at the cessation of a DBS-train . In one patient, where one could both record EMG and visually observe the onset of tremor at end of a DBS-train, these drops agree (within 1 sec.) with changes in EMG parameters and re-appearance of tremor, to allow prediction of onset of tremor . See Fig. 1. We also show that both LFP and raw sEMG allow discriminating tremor from voluntary movements (using wavelet parameters at certain bands) when DBS is OFF, as is essential for DBS control. While the results are preliminary, they point to the validity of using sEMG for control of DBS, at least in some ET patients
This article is published under license to BioMed Central Ltd. This is an open access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.